Picket Fence module documentation¶
The picket fence module is meant for analyzing EPID images where a “picket fence” MLC pattern has been made. Physicists regularly check MLC positioning through this test. This test can be done using film and one can “eyeball” it, but this is the 21st century and we have numerous ways of quantifying such data. This module attains to be one of them. It can load in an EPID dicom image (or superimpose multiple images) and determine the MLC peaks, error of each MLC pair to the picket, and give a few visual indicators for passing/warning/failing.
- Analyze either HD or regular MLCs - Just pass a flag and tell pylinac whether it’s HD or not.
- Easy-to-read pass/warn/fail overlay - Analysis gives you easy-to-read tools for determining the status of an MLC pair.
- Any Source-to-Image distance - Whatever your clinic uses as the SID for picket fence, pylinac can account for it.
- Account for panel translation - Have an off-CAX setup? No problem. Translate your EPID and pylinac knows.
- Account for panel sag - If your EPID sags at certain angles, just tell pylinac and the results will be shifted.
Although most terminology will be familiar to a clinical physicist, it is still helpful to be clear about what means what. A “picket” is the line formed by several MLC pairs all at the same position. There is usually some ideal gap between the MLCs, such as 0.5, 1, or 2 mm. An “MLC position” is, for pylinac’s purposes, the center of the FWHM of the peak formed by one MLC pair at one picket. Thus, one picket fence image may have anywhere between a few to a dozen pickets, formed by as few as 10 MLC pairs up to all 60 pairs.
Pylinac presents the analyzed image in such a way that allows for quick assessment; additionally, all elements atop
the image can optionally be turned off. Pylinac by default will plot the image, the determined MLC positions, two
“guard rails”, and a semi-transparent overlay over the entire MLC pair region. The guard rails are two lines parallel
to the fitted picket, offset by the tolerance passed to
analyze(). Thus, if a tolerance of 0.5 mm is passed, each
guard rail is 0.5 mm to the left and right of the invisible picket. Ideally, MLC positions will all be within these guard rails,
i.e. within tolerance, and will be colored blue. If they are outside the tolerance they are turned red.
If an “action tolerance” is also passed to
analyze(), MLC positions that are below tolerance but above the action
tolerance are turned magenta.
Additionally, pylinac provides a semi-transparent colored overlay so that an “all clear” or a “pair(s) failed” status is easily seen and not inadvertently overlooked. If any MLC position is outside the action tolerance or the absolute tolerance, the entire MLC pair area is colored the corresponding color. In this way, not every position needs be looked at. If all rows are green, then all positions passed.
Running the Demo¶
To run the picketfence demo, create a script or start in interpreter and input:
from pylinac import PicketFence PicketFence.run_demo()
Results will be printed to the console and a figure showing the analyzed picket fence image will pop up:
Picket Fence Results: 100.0% Passed Median Error: 0.062mm Max Error: 0.208mm on Picket: 3, Leaf: 22
Or plot the figure interactively: